Biphasic rectangular impulse with electrical mean equal zero (net zero DC).
Basic Unit Characteristics
|Weight:||350 g, 12.25 ounces|
|Length:||142 mm, 5.6 inches|
|Width:||99 mm, 3.9 inches|
|Height:||36 mm, 1.4 inches|
Max. output current per channel:
100 mA, for a max. charge of 2200 ohms
Max. quantity of electricity per impulse: 40 microcoulombs
Impulse width: 200 or 400 microseconds (200 for upper limbs and 400 for lower limbs)
Max. rise time for a 100 mA impulse: 2 microseconds
Maximum impulse frequencies: 120 Hz
Channels: Four independent channels with the possibility to individually regulate the current for each one. They are isolated one from the other and galvanically grounded.
Description of Accessories
Battery charger for recharging the battery:
Input 90-264 VAC / 47 to 63Hz / 0.25A max.
Four cables to link the four channels of the stimulator with the electrodes:
One blue cable, one green cable, one yellow cable, one red cable
4 small electrodes (5x5 cm, 2x2 inch)
4 large electrodes (5 x 10 cm, 2x4 inch)